If you are generating your clock internally, please contact National Instruments Technical Support. Error -200180Path between two switch channels is not available. To avoid the shutdown, ensure the device temperature does not get above the maximum recommended operating temperature. There's an example under labview 6.X\examples\daq\anlogin\anlogin.llb called Acquire&Proc Scans - SW Trig VI.
This is to prevent regeneration for output operations, or to reallocate resources for input operations." - If (iErr = -10804) Then NICode = "-10804 earlyStopError The transfer stopped prior to reaching Error -200223Specified threshold and hysteresis values for this channel create a triggering range that is not supported by your device. For example, a virtual channel using physical channel ao0 must be created before a virtual channel with physical channel ao1. Error -200006Record requested has been overwritten in the device memory. http://digital.ni.com/public.nsf/allkb/37752A79EB6A950886256C8B006E8870
Contact National Instruments if the problem persists. Error -200091Shared library version installed is incorrect. Error -200073SCXI module specified in the hardware configuration was not found.
The rate was coerced to the slowest possible convert rate. The requested sample was previously available, but has since been overwritten. Your data might be invalid. Labview Error 200279 Error -200015While writing to the buffer during a regeneration, the actual data generated might have alternated between old data and new data.
You must open the Configuration panel and associate the NI PXIe-5605 with a supported or external digitizer and a supported or external LO. Daqmx Error Codes Error -200018DAC conversion attempted before data to be converted was available. If you are using PCI devices, make sure they are connected with a RTSI cable and that the RTSI cable is registered in MAX. Cannot tristate these lines at this time.
Error -200183Switch channel names cannot be duplicated in the path string. Labview Custom Error Codes Change the data transfer mechanism from interrupts to DMA. 6. For acquisitions without a reference trigger, you can disallow buffer overwrites. Error -200137Initial Delay, High Time, and Low Time property values are inconsistent with one or more counter timebase properties.
You must open the Configuration panel and associate the NI PXI-5600 with a supported digitizer or an external digitizer. Enable memory mapping only when Data Transfer Mechanism is Programmed IO. Labview Error Code There are individuals who overreact when caught with this circumstance. Labview Error Daqmx There are two likely causes for this error: 1.
Error -200196Action at the end of the scan list is not valid for this device. Decrease the number of Pretrigger Samples per Channel, or increase Number of Samples per Channel. Reset it by doing one of the following: 1. All rights reserved. | United States MY ACCOUNT   INNOVATIONS SHOP SUPPORT COMMUNITY Home Community Home : Discussion Forums : Most Active Hardware Boards : Multifunction DAQ : Error NI DAQ Labview Visa Error -1073807339
Error -200009Timestamps have been overwritten. Error -200090Shared library was not found. The actual reference signal applied for calibration was different from the value you specified. Error -200152Data read from the EEPROM on the device is invalid.
Reconfigure the digital power-up states and perform a self-calibration. Labview Error Code 1 Error -200257Excitation property must be the same for related physical channels. If you are generating your clock internally, please contact National Instruments Technical Support.
Make sure the accessory specified in the hardware configuration is correct and that the hardware supports a built-in temperature sensor on this channel. Otherwise, make sure there is an available trigger line on the trigger bus shared between the devices. Error -200010Finite acquisition or generation has been stopped before the requested number of samples were acquired or generated. Labview Visa Error Codes Refer to Measurement & Automation Explorer for valid terminal names.
Reinstall NI-DAQmx, or download the latest version of the driver from the National Instruments website at ni.com. For detailed information about cabling, refer to documentation. Bring in your external sample clock through one of the higher-frequency front panel connectors, or use a lower sample rate. If your RAM is still workable, boosting the PageFile will also help.