The following constraint must hold: Offset < (Gain * Maximum Value / 2) Error -200270Interpolation rate specified is not possible for the given sample rate. On the other hand, another colleague here who has used LabView on a Mac, has been frustrated by it and stopped using it on Mac. Make sure you installed Microsoft Word correctly and create the report again. 1623 LabVIEW is unable to find the Microsoft Word components. The exception might have corrupted the LabVIEW memory. http://arxlabsolutions.com/error-code/oe-error-code-lg.php
Make sure that the channel is in the task or that the channel is a valid global channel. I'll have to keep a lookout for this in the future. Error -200069Device specified is not a valid switch device. I back up my files nightly, but I would like to avoid losing a day's worth of work as the result of a crash. - Should I consider doing my
Because of system and/or bus-bandwidth limitations, the driver could not write data to the device fast enough to keep up with the device output rate. To correct this error, load an instance of the polymorphic VI rather than the polymorphic VI itself. 1554 The current LabVIEW target cannot load control VIs. 1556 The reference is invalid. When scanning, you must use the original topology specified when the task was created. Error -200075Voltage data supplied is outside of the specified range.
Error -200035Clock rate specified is less than the minimum conversion rate of the ADC. The device will continue to function, but the accuracy of the measurements may be compromised. Error -200199Switch device must be reset before scanning. Labview Error 200279 Error -200221Amount of time allocated to perform this operation was exceeded.
To correct this error, ensure that the names you use for drag data are unique in each element of the data array for the method or event. 1385 Cannot start a If this error is returned from a Property or Invoke Node, the property or method might not be allowed for remote VI Server connections. Error -200148FREQOUT counter cannot generate the desired frequency. check over here Sync up changes before attempting this operation. 1500 If you obtain a user event reference in one application instance, you cannot use that user event reference in another application instance.
Error -200018DAC conversion attempted before data to be converted was available. Labview Custom Error Codes If you receive this error while trying to run File I/O VIs, refer to the KnowledgeBase at ni.com for more information. 2 Memory is full. ADC overrun errors are likely. Error -200012Specified physical channel does not support digital output.
Error -200126Lines on port C cannot be used for both handshaking control and static digital operations on an 8255 chip. http://zone.ni.com/reference/en-XX/help/371361K-01/lverror/misc_lv_error_codes/ Refer to the LabVIEW Help for a list of valid versions. 1526 This property does not apply when the plot legend of a graph or chart is in tree view. 1529 Labview Error Code To avoid the shutdown, ensure the device temperature does not get above the maximum recommended operating temperature. Labview Visa Error -1073807339 Would it be possible for you to post your code or a screenshot of your code to verify that this crash is reproducible on other Mac machines?
When a Boolean control is configured with a latching mechanical action, the Value property always returns an error. http://arxlabsolutions.com/error-code/ni-error-code.php The floating point data was outside of the range [-1.0, 1.0]. Or sign in with one of these services Sign in with Facebook Sign in with Twitter Sign in with LinkedIn Sign Up All Content All Content This Topic This Forum Advanced Error -200120Port is configured for input. Labview Error Daqmx
NATIVE INSTRUMENT SUPPORT COMMUNITY The Support Community is the new location where you can expect to receive assistance and feedback on technical support, sales or installation issues from our Support Agents This is not allowed when waiting for devices to settle. The floating point data was outside of the range [-1.0, 1.0]. Source A connection separator, sequence separator, or valid switch action terminator must follow a switch action.
Get the Number of Possibly Invalid Samples property to see how many samples might be invalid. Labview Visa Error Codes If you are generating your clock internally, please contact National Instruments Technical Support. Solution: This error occurs when there are multiple different driver versions on the system, perhaps as a result of an upgrade (e.g.
Error -200064The NI PXIe-5663 is not configured properly. Void can be used only as return type of function. 1186 Cannot show or hide the label on its own. Error -200281Reading relative to the reference trigger or relative to the start of pretrigger samples position before the acquisition is complete. Labview Error 50103 Error -200211Multiple relay names were specified for a single relay operation.
The device driver needed to execute the In Port and Out Port VIs is not present. For example, this error might occur when a front panel terminal is inside a structure. 1430 The path is empty or relative. This is caused by trying to allocate a buffer that is too large for memory. have a peek here This document has been archived and is no longer updated by National Instruments Note: This page is about NI-DAQ also known as Traditional NI-DAQ (Legacy).
No reference could be returned. Error -200191Task was created with a topology different from the current topology. Change the range or the data. Error -200229Buffer is too small to fit read data.
If you have re-connected a removable device, you may need to restart LabVIEW to use the device again. 29 Print aborted. 30 Bad print record. 31 Print driver error. 32 Operating Refer to the documentation for more information. Double-click the VI or function and replace the general error handling with an error out indicator or some other form of error handling. Error -200043Hardware necessary for this route is in use by another task or tasks.
To correct this error, ensure that a drag and drop operation is in progress when you call this function and that the data type and drag data name match what is Guitar Rig has been one of the most frustrating experiences I've had to deal with in the last while. To correct this error and inline the subVI, remove the implicit Property Node or Invoke Node. 1488 The migration file cannot load because it is no longer supported. 1489 The migration The specified wave format is not supported.
Error -200192Switch device supports continuous scanning only. This file cannot be saved until all dependent files have been named. 1451 One or more untitled dependencies exist. Cannot configure this port or any of its lines for handshaking at this time. A VI item in the palette data array is not supported in this version of LabVIEW.
If you are using an external sample clock or an external reference clock, make sure it is connected and within the jitter and voltage specifications. Increase the number of samples you read each time you invoke a read operation. 3. Error -200044Invalid enumeration value was encountered during export. Accessing the VI Name using the Property Node appears to work no problem on the RT, though. Does anyone know what could be causing the problem?
The palette type input is invalid. Refer to the documentation for information on connection ranges and semicolon modifiers. Therefore, you cannot use lines 0, 1, 2, and 4 for regular digital I/O. Error -200067The NI PXIe-5665 (3.6 GHz) is not configured properly.