Some of the examples that used these VIs could not be opened in LabVIEW 5.1. Support for fetching data in LabVIEW 7.0 in the Waveform Data Type (WDT). NI-SCOPE 2.1.4 fixed the following errors with NI-SCOPE 2.1.3: Some of the read VIs were saved in LabVIEW version 6.0 instead of 5.1. NI-SCOPE is IVI-Scope compliant, meaning that software written to the NI-SCOPE interface will be completely compatible with future products in the NI High-Speed Digitizer family. http://arxlabsolutions.com/error-code/nrg-error-codes.php
Continuous acquisition for NI 5112 and NI 5620. Please try the request again. Auto Setup The Auto Setup function can be used to automatically set the vertical range, sample rate, minimum record length, and the trigger level. Kontakt Rechtliche Hinweise | Datenschutz | © National Instruments Corporation. http://digital.ni.com/public.nsf/allkb/FD9FEFCDD2F95C29862572AA006AC909
The Fetch functions are easier to use and have new functionality Allows fetching multiple channels Allows fetching multiple records Returns a cluster suitable for wiring directly to charts or graphs in With regards, Niranjan.Ch simple program.vi 24 KB Error.JPG 85 KB 0 Kudos Message 1 of 5 (3,768 Views) Reply 0 Kudos Re: NI Scope Error-1074126845 The_Devil Member 12-09-2010 10:17 PM Options Step 3: Acquire Data After the digitizer is configured for your application, you may start the acquisition. D.
If you are acquiring large blocks of data or if there is a delay on the trigger, the niScope Read can execute before the acquisition started by the niScope Initiate is instrument handle is the handle that you obtain from niScope Initialize that identifies a particular instrument's session popup message enables a pop up message when the case is set to Yes For a more thorough explanation of the NI-SCOPE instrument driver and more information on programming with NI-SCOPE, refer to the NI High-Speed Digitizers Help. Labview Visa Error Codes The system returned: (22) Invalid argument The remote host or network may be down.
Putting it all Together The following diagram illustrates a flow diagram of a typical acquisition. New API functions including Commit, Reset Device, Disable and firmware revision through Revision Query Simulation for all devices supported by NI-SCOPE. When i am running the code the following error is comming (Error-1074126845 occured at niScope Multi Fetch WTD) the description of error is maximum time exceeded before the operation completed. look at this site Therefore, the error cluster must be connected to all the VIs before the error message is displayed.
Your cache administrator is webmaster. Labview Error 200279 However, on a single record acquisition, the fetch functions do not wait for the trigger holdoff to expire. i am facing one problem . NI-SCOPE 2.1.3 added the following functionality to NI-SCOPE 2.1.2: Support for the NI 5621 for PXI.
Another benefit is the elimination of ordering dependencies of configuration calls which simplifies user programs. http://digital.ni.com/public.nsf/allkb/B255EB105B299240862577380071314B Known Issues Link Library NI-SCOPE 2.1.4 and previous used to install a duplicate version of the NI-SCOPE C link library under the name of "niscope_32.lib". Labview Error Code niScope Initialize.vi C Function: niScope_Init Initialize sets the NI-SCOPE driver and the digitizer to a known state. Labview Visa Error -1073807339 Cheers.!! 0 Kudos Message 2 of 5 (3,748 Views) Reply 0 Kudos Re: NI Scope Error-1074126845 Niranjan.Chell… Member 12-10-2010 12:08 AM Options Mark as New Bookmark Subscribe Subscribe to RSS Feed
Poor|Excellent Yes No Document Quality? Check This Out This behavior was fixed to properly return an error message. My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation. B. Labview User Error Codes
If you have modified any old examples these will be overwritten by the new versions. This VI takes the error code returned by NI-SCOPE VIs and returns the interpretation as a user-readable string. It is the first instrument driver function called and establishes communication with the instrument. http://arxlabsolutions.com/error-code/ni-error-codes.php If you change the rate of the source before setting the attribute, you may get an error.
However, software written to the NI-SCOPE interface will be completely compatible with future products in the NI high-speed digitizer family. Labview Error 1073807339 Updated IVI Configuration Store behavior The IVI Configuration Store file format now requires that all names be case-sensitive. Below is a model structure of a program written with an instrument driver.
Instrument Driver Programming Basics What is an instrument driver? Select Modify on the NI-SCOPE entry and remove the SCOPE Soft Front Panel feature. Step 2: Configure After you have opened a session with the high-speed digitizer, you need to configure the hardware settings. Labview Error Visa Read Primary Software: Primary Software Version: 3.6 Primary Software Fixed Version: N/A Secondary Software: N/A Problem: When executing my NI Scope application I get:Error -1074118647 occurred at niScope Multi Read Cluster.vi:1Possible reason(s):A
Using NI-SCOPE Call By Reference in LabVIEW Call By Reference is a LabVIEW feature that allows the execution source for a VI to be dynamically determined at run time. NI-SCOPE 2.0 occasionally failed to report an error for a measurement if a previous measurement had failed. 5. What is happening is that both the niScope Initiate and the niScope Read are attempting to put the digitizer into a state to fetch data. have a peek here Visual Basic users should declare variables of type String.
The installer is a self-extracting ZIP file. Add Comments 1 2 3 4 5 My Profile|Privacy|Legal|Contact NI© 2009 National Instruments Corporation. Array Measurements Array measurements are measurements on waveforms that result in an array of values such as FFT amplitude spectrum. Windows examples Autosetuped acquisition In this example by calling niScope_AutoSetup driver automatically try to setup acquisition parameters.
To avoid having the NI-SCOPE DLL unload after each Call By Reference VI call, statically place any NI-SCOPE VI, such as niScope_Initialize.vi on the diagram inside a case structure wired to An instrument driver is a set of software routines that control a programmable instrument. Call niScope_AcquisitionStatus to find out when the entire acquisition including trigger holdoff has completed. The default maximum input frequency is now 35Mhz for this device.
Programs linking against the duplicate name need to be changed to link against "niscope.lib". The number of Configuration functions depends on the complexity of the instrument. Poor|Excellent Yes No Document Quality? New external clocking example for LabVIEW.